AGILENT E4440A MANUAL PDF
This manual provides documentation for the following instruments: EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3. This manual provides documentation for the following instruments: PSA Series. EA (3 Hz – GHz). EA (3 Hz – GHz). EA (3 Hz – GHz). Agilent Technologies. PSA Series Spectrum Analyzer. This manual provides documentation for the following instruments: EA (3 Hz – GHz). EA (3.
|Genre:||Health and Food|
|Published (Last):||3 August 2017|
|PDF File Size:||11.81 Mb|
|ePub File Size:||1.42 Mb|
|Price:||Free* [*Free Regsitration Required]|
Options see page Unless specified otherwise, all options are available when you order a spectrum analyzer; some options are also available as kits that you can order and install after you receive the analyzer.
Agilent Technologies E4440A Getting Started Manual
It is effective over a frequency range of dc to 2 GHz. Options and Accessories Ordering Options and Accessories Ordering Options and Accessories Options and accessories help you configure the analyzer for your specific applications, and enable you to use the analyzer.
If a problem persists, you may choose to: Don’t show me this message again. If your analyzer is covered by a separate maintenance agreement, please be familiar with aggilent terms.
Each softkey More has several characters on it. Making a Basic Measurement This chapter provides information on basic analyzer operation. Arrow Keys Numeric Keypad Enters a specific value.
E4440A PSA Spectrum Analyzer, 3 Hz to 26.5 GHz
Make sure there is no disk in the front-panel disk drive. For more information on making measurements, see the measurement guide for your analyzer. Place a marker labeled 1 on the 10 MHz peak, as shown in Press Peak Search Note that the frequency and amplitude of the marker appear both in the active function block, and in the upper-right corner of the screen.
AC Power Cords Chapter Is the analyzer turned on? Agilent Technologies shall not be liable for errors contained herein or for incidental or consequential damages in connection with the furnishing, performance, or use of this material.
A pound uncoupled function. Using The Front Panel Using the Front Panel Entering Data When setting measurement parameters, there are several ways to enter or modify the value of the active function: To access the Save menu, press, 2.
Making a Basic Measurement This chapter is divided into the following sections: If the printer is ready and the printer setup was successful, a printout of the analyzer display will be printed. Then press the desired terminator either a unit softkey, or the Terminate a value that requires a unit-of-measurement.
Please read it and become familiar with its terms. User Preset This type of preset restores the currently selected mode to a known Mode Preset Front and Rear Panel Features This chapter gives you an overview of the front e44440a rear panels of your analyzer.
Firmware Revision To view the firmware revision of your analyzer, press call Agilent Technologies regarding your analyzer, it is helpful to have this revision and the analyzer serial number available.
To access the menu, press Rename 2. This symbol is used to indicate the OFF position of the power button. In this example you are loading a trace file; RS This feature not fully implemented; to be implemented in a future release. Directories are manusl as DIR and drives have nothing listed under Type:.
Agilent Technologies makes no warranty of any kind with regard to this material, including but not limited to, the implied warranties of merchantability and fitness for a particular purpose. Firmware Revision Firmware Revision To view the firmware revision agilenh your analyzer, press call Agilent Technologies regarding your analyzer, it is helpful to have this revision and the analyzer serial number available.
Sales and Service Office and are listed below. Viewing Traces 1, 2, and 3 Step 2. To be implemented in future firmware upgrade.
AGILENT TECHNOLOGIES EA GETTING STARTED MANUAL Pdf Download.
agileny EA 3 Hz – The analyzer has three types of preset: While not all of the zeros following the decimal in the active function block are significant, the numbers after the decimal in the marker annotation upper-right corner of screen are significant. In Case of Difficulty This chapter includes information on how to check for a problem with your Agilent Technologies spectrum analyzer, and how to return it for service.
Select the type of file you want to rename setup, state, manyal, limits, screens, or corrections. WARNING If this product is to be energized via an external auto transformer for voltage reduction, make sure that its common terminal is connected to a neutral earthed pole of the power supply.
Viewing a Signal 1. Ordering Options And Accessories Options and Accessories Ordering Options and Accessories Ordering Options and Accessories Options and accessories help you configure the analyzer for your specific applications, and enable you qgilent use the analyzer.
Page 34 Making a Basic Measurement This chapter is divided into the following sections: Increments or decrements the current value. Factory Preset Restores the analyzer to a user-defined state. Improving Frequency Accuracy Preselection reduces mixer overload from broadband signals, reduces radiation of local oscillator harmonics back to the device under test, and reduces the level of image and multiple responses displayed.